Parameters for C:\MassLynx\DEFAULT.PRO\ACQUDB\posneg_1000.EXP
Created by Masslynx V4.1
 
 
Use TTP 4GHz TDC	YES
Dynamic Range Enhancement	NO
Pirani Pressure(mbar)	2.07e0
Penning Pressure (mbar)	6.03e-7
 
Instrument Parameters - Function 1:
Polarity	ES+
Analyser	V Mode
Capillary (V)	3000.0
Sample Cone (V)	100.0
Desolvation Temp (C)	150.0
Source Temp (C)	100.0
Cone Gas Flow	50.0
Desolvation Gas Flow	300.0
Syringe Type	
Ion Guide One	5.0
Aperture 1 Voltage	15.0
Ion Energy (V)	105.0
Aperture 2 Voltage	4.0
Hexapole DC Voltage	6.0
Aperture 3 Voltage	4.0
Acceleration (V)	200.0
Y Focus (V)	0.0
Steering (V)	0.0
Tube Lens (V)	150.0
Attenuated Z Focus (V)	500.0
Normal Z Focus (V)	60.0
TOF Flight Tube (V)	5630.0
Reflectron (V)	1780.0
Pusher Voltage	800.0
Pusher Offset Voltage	0.00
Puller Voltage	740.0
Puller Offset Voltage	0.00
MCP Detector (V)	2400.0
Pusher Cycle Time	Auto (34.0)
Pusher Frequency	29411.76
Pusher Width	4.00
Centroid Threshold	1.0
Min Points	2.0
Np Multiplier	0.70
Resolution	6500.0
Lteff	1081.0000
Veff	5673.7690
Trigger Threshold (mV)	600.0000
Signal Threshold (mV)	20.0000
Data Threshold	0.0000
DXC Temperature	25.0
IonGuide1InitialRF	150.0
IonGuide1FinalRF	150.0
IonGuide2InitialRF	200.0
IonGuide2FinalRF	200.0
Fixed Hexapole RF	False
HexOffset1	0.0
HexOffset2	0.0
HexOffset3	0.0
HexOffset4	0.0
HexFactor1	2.0
HexFactor2	2.0
HexFactor3	2.0
HexFactor4	2.0
DRE Mass 0.0000 Setting 2.2000
DRE Mass 280.0000 Setting 22.0000
DRE Mass 1000.0000 Setting 22.0000
DRE Mass 2000.0000 Setting 22.0000
DRE Mass 3000.0000 Setting 22.0000
 
 
Instrument Parameters - Function 2:
Polarity	ES-
Analyser	V Mode
Capillary (V)	3000.0
Sample Cone (V)	100.0
Desolvation Temp (C)	150.0
Source Temp (C)	100.0
Cone Gas Flow	25.0
Desolvation Gas Flow	400.0
Syringe Type	
Ion Guide One	5.0
Aperture 1 Voltage	15.0
Ion Energy (V)	105.0
Aperture 2 Voltage	4.0
Hexapole DC Voltage	6.0
Aperture 3 Voltage	4.0
Acceleration (V)	200.0
Y Focus (V)	0.0
Steering (V)	0.0
Tube Lens (V)	150.0
Attenuated Z Focus (V)	500.0
Normal Z Focus (V)	60.0
TOF Flight Tube (V)	5630.0
Reflectron (V)	1780.0
Pusher Voltage	800.0
Pusher Offset Voltage	0.00
Puller Voltage	740.0
Puller Offset Voltage	0.00
MCP Detector (V)	2600.0
Pusher Cycle Time	Auto (34.0)
Pusher Frequency	29411.76
Pusher Width	4.00
Centroid Threshold	1.0
Min Points	2.0
Np Multiplier	0.70
Resolution	5500.0
Lteff	1081.0000
Veff	5671.3833
Trigger Threshold (mV)	600.0000
Signal Threshold (mV)	20.0000
Data Threshold	0.0000
DXC Temperature	25.0
IonGuide1InitialRF	150.0
IonGuide1FinalRF	150.0
IonGuide2InitialRF	200.0
IonGuide2FinalRF	200.0
Fixed Hexapole RF	False
HexOffset1	0.0
HexOffset2	0.0
HexOffset3	0.0
HexOffset4	0.0
HexFactor1	2.0
HexFactor2	2.0
HexFactor3	2.0
HexFactor4	2.0
DRE Mass 99999.9000 Setting 50.0000
DRE Mass 99999.9000 Setting 50.0000
DRE Mass 99999.9000 Setting 50.0000
DRE Mass 99999.9000 Setting 50.0000
DRE Mass 99999.9000 Setting 50.0000
 
 
